It is common for forensic practitioners to calculate an individual's likely blood alcohol concentration following the consumption of alcoholic beverage(s) for legal purposes, such as in driving under the influence (DUI) cases. It is important in these cases to be able to give the uncertainty of measurement on any calculated result, for this reason uncertainty data for the variables used for any calculation are required. In order to determine the uncertainty associated with the alcohol concentration of beer in the UK the alcohol concentration (%v/v) of 218 packaged beers (112 with an alcohol concentration of ≤5.5%v/v and 106 with an alcohol concentration of >5.5%v/v) were tested using an industry standard near infra-red (NIR) analyser. The range of labelled beer alcohol by volume (ABV's) tested was 3.4%v/v – 14%v/v. The beers were obtained from a range of outlets throughout the UK over a period of 12 months. The root mean square error (RMSE) was found to be ±0.43%v/v (beers with declared %ABV of ≤5.5%v/v) and ±0.53%v/v (beers with declared %ABV of >5.5%v/v) the RMSE for all beers was ±0.48%v/v. The standard deviation from the declared %ABV is larger than those previously utilised for uncertainty calculations and illustrates the importance of appropriate experimental data for use in the determination of uncertainty in forensic calculations.
|Number of pages||5|
|Journal||Science and Justice|
|Early online date||6 Mar 2018|
|Publication status||Published - Jul 2018|
- Alcohol technical defence
- Blood alcohol calculation
- Alcohol by volume
- Driving under the influence
FingerprintDive into the research topics of 'The influence of alcohol content variation in UK packaged beers on the uncertainty of calculations using the Widmark equation'. Together they form a unique fingerprint.
Dawn Louise Maskell
- School of Engineering & Physical Sciences - Professor
- School of Engineering & Physical Sciences, Institute of Biological Chemistry, Biophysics and Bioengineering - Professor
Person: Academic (Research & Teaching)