axis of the spectrograph. It is anticipated that the size of a single-object spectrograph fed with light in this manner would be essentially independent of the telescope aperture size. A further anticipated benefit is that such spectrographs would be free of ‘modal noise’, a phenomenon that occurs in high-resolution multimode fibre-fed spectrographs due to the coherent nature
of the telescope point-spread-function (PSF). We seek to address these aspects by integrating a multicore fibre photonic lantern with an ultrafast laser inscribed three-dimensional waveguide interconnect to spatially reformat the modes within the PSF into a diffraction-limited pseudo-slit. Using the CANARY adaptive optics (AO) demonstrator on the William Herschel Telescope, and 1530 80 nm stellar light, the device exhibits a transmission of 47 – 53 per cent depending upon the mode of AO correction applied.We also show the advantage of using AO to couple light into such a device by sampling only the core of the CANARY PSF. This result underscores the possibility that a fully-optimised guided-wave device can be used with AO to provide efficient spectroscopy at high spectral resolution.
- instrumentation: adaptive optics
- instrumentation: spectrographs
ASJC Scopus subject areas
- Physics and Astronomy(all)